Human Magnetoencephalography Measurement by Highly Sensitive SERF Atomic Magnetometer
【Abstract】A highly sensitive noncryogenic rubidium magnetometer based on spin exchange relaxation free (SERF) is designed, whose sensitivity at 15 Hz isWith this SERF magnetometer, the difference in the human brain magnetic field induced by eye opening and closing is recorded inside the shielding barrel. This SERF magnetometer is operated in double light mode with a pump-probe arrangement. Compared with single beam arrangement, this SERF magnetometer can achieve a higher sensitivity and does not require any extra magnetic modulation. Thus, the complexity of the acquisition system is reduced and the lock-in amplifier is not needed any more. Moreover, this kind of configuration is easily adapted to miniaturize the sensor array for the future whole-head magnetoencephalography equipment.
【Keywords】 measurement; atomic magnetometer; spin exchange relaxation free; magnetoencephalography;
 Cheyne D, Bostan A C, Gaetz W, et al. Eventrelated beamforming: a robust method for presurgical functional mapping using MEG [J]. Clinical Neurophysiology, 2007, 118 (8): 1691–1704.
 Colon A J, Ossenblok P, Nieuwenhuis L, et al. Use of routine MEG in the primary diagnostic process of epilepsy [J]. Journal of Clinical Neurophysiology, 2009, 26 (5): 326–332.
 Gratta C D, Pizzella V, Tecchio F, et al. Magnetoencephalography: a noninvasive brain imaging method with 1 ms time resolution [J]. Reports on Progress in Physics, 2001, 64 (12): 1759–1814.
 Cohen D, Cuffin B N. Demonstration of useful differences between magnetoencephalogram and electroencephalogram [J]. Electroencephalography and Clinical Neurophysiology, 1983, 56 (1): 38–51.
 Hamalainen M, Hari R, Ilmoniemi R J, et al. Magnetoencephalography: theory, instrumentation, and applications to noninvasive studies of the working human brain [J]. Reviews of Modern Physics, 1993, 65 (2): 413–497.
 Cohen D. Magnetoencephalography: detection of the brain′s electrical activity with a superconducting magnetometer [J]. Science, 1972, 175 (4022): 664–666.
 Liang S Q, Yang G Q, Xu Y F, et al. Simultaneously improving the sensitivity and absolute accuracy of CPT magnetometer [J]. Optics Express, 2014, 22 (6): 6837–6843.
 Budker D, Gawlik W, Kimball D F, et al. Resonant nonlinear magneto-optical effects in atoms [J]. Physical Review, 2002, 74 (4): 1153–1201.
 Yang A L, Yang G Q, Cai X M, et al. A laser pump-re-pump atomic magnetometer [J]. Chinese Physics B, 2013, 22 (12): 120702.
 Happer W, Tang H. Spin-exchange shift and narrowing of magnetic resonance lines in optically pumped alkali vapors [J]. Physical Review Letters, 1973, 31 (5): 273–276.
 Happer W, Tang H. Effect of rapid spin exchange on the magnetic-resonance spectrum of alkali vapors [J]. Physical Review A, 1977, 16 (5): 1877–1891.
 Dang H B, Maloof A C, Romalis M V. Ultrahigh sensitivity magnetic field and magnetization measurements with an atomic magnetometer [J]. Applied Physics Letters, 2010, 97 (15): 151110.
 Sander T H, Preusser J, Mhaskar R, et al. Magnetoencephalography with a chip-scale atomic magnetometer [J]. Biomedical Optics Express, 2012, 3 (5): 981–990.
 Xia H, Ben-Amar Baranga A, Hoffman D, et al. Magnetoencephalography with an atomic magnetometer [J]. Applied Physics Letters, 2006, 89 (21): 211104.
 Johnson C, Schwindt P D D, Weisend M. Magnetoencephalography with a two-color pumpprobe, fiber-coupled atomic magnetometer [J]. Applied Physics Letters, 2010, 97 (24): 243703.
 Kamada K, Sato D, Ito Y, et al. Human magnetoencephalogram measurements using newly developed compact module of high-sensitivity atomic magnetometer [J]. Japanese Journal of Applied Physics, 2015, 54 (2): 026601.
 Colombo A P, Carter T R, Borna A, et al. Fourchannel optically pumped atomic magnetometer for magnetoencephalography [J]. Optics Express, 2016, 24 (14): 15403–15416.
 Boto E, Holmes N, Leggett J, et al. Moving magnetoencephalography towards real-world applications with a wearable system [J]. Nature, 2018, 555 (7698): 657–661.
 Sheng J W, Wan S G, Sun Y F, et al. Magnetoencephalography with a Cs-based high sensitivity compact atomic magnetometer [J]. Review of Scientific Instruments, 2017, 88 (9): 094304.
 Allred J C, Lyman R N, Kornack T W, et al. High sensitivity atomic magnetometer unaffected by spinexchange relaxation [J]. Physical Review Letters, 2002, 89 (13): 130801.
 Kornack T W, Smullin S J, Lee S K, et al. A lownoise ferrite magnetic shield [J]. Applied Physics Letters, 2007, 90 (22): 223501.
 Smullin S J, Savukov I M, Vasilakis G, et al. Lownoise high-density alkali-metal scalar magnetometer [J]. Physical Review A, 2009, 80 (33): 033420.
 Ledbetter M P, Savukov I M, Acosta V M, et al. Spin-exchange-relaxation-free magnetometry with Cs vapor [J]. Physical Review A, 2008, 77 (3): 033408.
 Kominis I K, Kornack T W, Allred J C, et al. Asubfemtotesla multichannel atomic magnetometer [J]. Nature, 2003, 422 (6932): 596–599.
 Hansen P C, Kringelbach M L, Salmelin R. MEG: an introduction to methods [M]. Oxford: Oxford University Press, 2010: 35, 169.