(2.中国计量科学研究院, 北京 100013)
【摘要】针对高精度复合式测量机中的激光传感器进行了高精度的标定, 提出了利用已知底角的双斜面标定块和正弦定理标定光轴垂直度误差偏角的方法, 分析了影响标定精度的因素。该方法避免了传感器光轴与工作台不垂直引入的各个方向的误差, 可有效地提高该传感器的测量精度, 为更高精度的多传感器复合测量提供了基础。
【基金资助】 国家重大科学仪器设备开发专项 (2013YQ17053903) ;
Calibration Method for Optical Axis Perpendicular Error of Laser Sensor
(2.National Institute of Metrology of China, Beijing, China 100013)
【Abstract】The high-precision calibration of laser sensors in high-precision composite measuring machines is conducted. The calibration method of the optical axis perpendicular error angle by using the calibration block of double slopes with known base angles and the sine theorem is proposed. This method can avoid the errors from various directions caused by the non-perpendicular orientation between the optical axis of sensors and the worktable, which can effectively improve the measurement accuracy of this sensor and provide a basis for multi-sensor duplex measurement with higher accuracy.
【Keywords】 sensors; calibration of perpendicular error; fitted plane slope; optical axis; high-precision complex measurement;
【Funds】 National Key Foundation for Exploring Scientific Instrument Program (2013YQ17053903) ;
 Singh P, Wu Y, Kaucic R, et al. Multimodal industrial inspection and analysis [J]. Journal of Computing & Information Science in Engineering, 2007, 7 (1): 102–107.
 Waurzyniak P. Multisensor systems ramp up-Faster processors, improved accuracy spur multisensor adoption [J]. Manufacturing Engineering, 2003, 130 (6): 61–68.
 Yang X G, Huang Y M, Gao F, et al. Research on calibration algorithm of analogue touch probe with nonorthogonal coordinate system [J]. Chinese Journal of Scientific Instrument, 2007, 28 (10): 1826–1830 (in Chinese).
 Zhu C G, Huang M Q. Calibration of Master Ball on Coordinate Measuring Machine [J]. Shanghai Measurement and Testing, 1995 (5): 25–27 (in Chinese).
 Luo Z. Calibration and Correct Use of 3D Trigger Type Probe [J]. Tool Engineering, 1992 (5): 38–40 (in Chinese).
 Zhang G X, Zhang D F. The optical plan for measuring center and radius of a ball part by 4-point method [J]. Acta Metrologica Sinica, 1993, 14 (4): 247–250 (in Chinese).
 Zhang Z. Flexible camera calibration by viewing aplane from unknown orientations [C]. Proceedings of the 7th International Conference on Computer Vision, 1999: 6365272.
 Zhang Z. Camera calibration with one-dimensional objects [J]. IEEE Transactions on Pattern Analysis & Machine Intelligence, 2004, 26 (7): 892–899.
 Kong W Q, Liu J N, Da F P, et al. Calibration method based on general imaging model for micro-object measurement system [J]. Acta Optica Sinica, 2016, 36 (9): 0912003 (in Chinese).
 Lai H X, Zhang Z Y, Zhu L, et al. Nonlinear distortion correction of camera based on linear characteristic [J]. Laser & Optoelectronics Progress, 2016, 53 (2): 021502 (in Chinese).
 Fang J W, Zhang G X, Qiu Z R. Calibration system for surveying axis of non-contact laser probe [J]. Electronic Measurement Technology, 2006, 29 (3): 160–161 (in Chinese).
 Lu K Q, Wang W, Chen Z C. Calibration of laser beam-direction for point laser sensors [J]. Optics and Precision Engineering, 2010, 18 (3): 880–886 (in Chinese).
 Zhou H C, Zeng L Z, Chen J H, et al. Calibration of light beam′s direction of point light source probe [J]. Chinese Journal of Scientific Instrument, 2004, 25 (3): 388–391 (in Chinese).
 Zhang J H, Tan J B. Theoretical analysis of surface slope measurement with a confocal microscope [J]. Journal of Optoelectronics·Laser, 2002, 13 (12): 1272–1275 (in Chinese).
 National standardization technical committee. Angle block gauge GB3/T 22521-2008 [S]. Beijing: China Standard Press, 2009.